Release 2022-03-14 (Version 18.104.22.1684)
Analyses approve more measurements (by adjusted parameters) than before:
- Eliminates Glucose blank and calibration problems
- Eliminates low concentration analyses issues
- An entry in the service log reveals values that are approved by the adjusted parameters
Control Samples LOT number gets stuck (2022-02-09)
In order to successfully cope with Control Samples nominal control levels a new Service Tool has been released. Especially for lot T27524, if Nominal Control Levels have been updated using a nominalcontrollevels.ini file, the lot number gets stuck and can’t be updated. To release the lot T27524 so i may be updated:
Download the file below, extract the zip-file to an empty USB stick and use it as a Service Tool and start UpdateDB according to the instructions below.
Lactate Calibration ResponseMax Fix (2021-11-11)
In order to cope with sensitive Lactate reagent batches a new Service Tool has been released. Download the file below, extract the zip-file to an empty USB stick and use it as a Service Tool to update the ResponseMax limit for Lactate calibrations.